The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||19 November 2007|
|PDF File Size:||7.70 Mb|
|ePub File Size:||8.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Ohmmeter Accuracy Test After the first measurement cycle is completed, the delayed trigger will iniate a second measurement cycle.
G and G 2 are the circuit gains of the particu- lar measurements, and fCr factor. Off 1 High Ret.
3455A 5 or 6-Digit Digital Multimeter/Voltmeter
This transfers the number from the storage register specified to the display. However, the device must be HP-IB compatible. Image not available Photos not available for this variation.
Used with field wiring terminals to in- dicate the terminal which must be connected to ground before opierating equipment. Uprange the dc standard to the V range. Jumper Removal A15 Board The following is an explanation of the various times involved in a hhp se- quence. In any correspondence, identify the instrument by model number and full serial number.
Tlic use of repaired fuses and the short-circuiting of fuse holders must be avoided. Assemble and Test Point Loading 8-H This action takes place when an input measurement is taken. msnual
Ohms Reference Measurement Turn-On and Warm-Up Open the input leads and connect the unknown resistor to the leads. All tests can be performed without access to the interior of the instrument.
The error equation becomes: This allows common-mode current to flow through lead resistance Rb causing some measurement error. A reading is taken on all intermediate non-optimum ranges until the correct range hpp found.
However, Buyer manuwl pay all shipping charges, duties, and taxes for products returned to -hp- from another country. Derive 6-month test limits from AC Accuracy specifications listed in Table AC Convertor Troubleshooting 8-C For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
HP Hewlett Packard a Digital Voltmeter Gold Board Vintage. | eBay
The square root amplifier reverses the action of squaring amplifier. Derive 6-month test limits fromOhms Accuracy specifications listed in Table The examples in the Appendix can be helpful when you write programs for the 34SSA.
Wrap instrument in heavy paper or plastic before placing in an inner container.
Bus Messages This completes the dc common-mode Rejection test. The self test will be repeated until another function is selected.
Hewlett Packard HP 3455a 6-digit Digital Voltmeter
The A normally leaves the factory with the switch set to listen address 6 and talk address V decimal code The 34S5A reading should be within Accepi Data — IA addressed to listen or ATN true psec per character typical 19 delay tourcel Output Data – A addressed to talk Atec per character typical 10 delay acceptor Gefieral Auto Cal must be on for 75 seconds to meet all specifications Overload Irtdication: Ohms Offsets 8-D-ll.
Example C is similar majual A with the ex- ception that connecting guard in this manner allows any common-mode current generated between the source low and powerline ground to flow in the measurement circuit.
When the A is the External Trigger mode, the user can trigger the instrument from an external trigger pulse. The measurement time on 345a range should be added to the total measurement time. Each set of specifications includes an accuracy specification for each voltage or ohms range.