Download HICKOK A TUBE TESTER SCH service manual & repair info for electronics experts. To my knowledge there is no written calibration procedure by Hickok for the A . Calibration was set during manufacture by the use of mostly. Hickok tube tester model A. Model & A & most other model manuals are available with high quality schematics.
|Published (Last):||23 May 2015|
|PDF File Size:||19.46 Mb|
|ePub File Size:||14.55 Mb|
|Price:||Free* [*Free Regsitration Required]|
Learn more – opens in new window or tab.
Jun Sun 07, See other items More Specific Procedures to test early C- series tube. If someone thinks you are pestering them with questions then they can ignore you. I guess I am still on the hunt to have the most complete tester, but I’ve mellowed out since my Original post. About the best you can do is check the filament with an ohmmeter.
I genuinely hope he is OK. I think that the self-bias binding post may be a good place for an ammeter, but does not work on the 2nd half of twin triode – only good for section one. Used for testing rectifiers and radio tubes. Seller assumes all responsibility for this listing. The ’00 and ’00A are tricky tubes to test. Hickok Electrical Instrument Co.
Hickok /A Tube Tester Calibration Procedure Test Data & Owners Manual CDrom | eBay
It tells how to set the tester up and what the voltages should be for the plate, screen, bias, signal, filament and leakage circuit. Seller information elderlyelectronics99 This amount is subject to change until you make payment. The missing page s Manula see are between these tubes: Mouse over to Zoom – Click to enlarge.
Good Luck and let me know if you find that manual info. One common file is “settings. AZStuff, I’ve temporarily posted the whole pdf file here and you can see for yourself: Data is extracted verbatim from the Hickok Instruction Manual No. Minimum monthly payments are required. I got pretty much the same way before my first tube tester purchase.
Dynamic Mutual Conductance Tube Tester 752A
Thank you for your time while visiting our auctions. Hickok tube tester theories, a description of Hickok adapters and dealer pricing from The filament voltage selection on the A tester offers only The A calls for manuaal and Each section of the tube is independently tested.
The outstanding feature of the Model A is the “dual-tube test” which permits the testing of tubes containing electrically similar sections with one setting of the selector switches. Our main goal is customer satisfaction and to help you revive that special piece of equipment with quality tubes, parts, and manuals.
Oct 10, Coweta, Oklahoma, United States. The Introduction Chapter I describes the model A as employing the. I checked bias and bias 22 but was not sure what spec to go by. Instruction manual identifies the case as: Tubetechie, Don’t be embarrassed or afraid to ask questions here.
Jun Sun 07, 7: So about 4, tubes. I have the test data pages you are missing. See terms – manyal in a new window or tab.
Subject to credit approval. Use red slider bar for more.
The item you’ve selected was not added to your cart.